The alias theorems: practical undersampling for expert engineers
Physicist and engineer Harry Nyquist's 1928 paper on telegraphtransmission theory revealed that complete reconstruction of an Nelement signal is possible if you know N/2 sinusoidal components (Reference 1). This theory developed into Nyquist's sampling theorem, which states that complete reconstruction of a waveform is possible from samples taken at a rate greater than twice the highest frequencyharmonic component. If you sample the signal more slowly, an alias results, and information is lost.
The alias is unknown to novice engineers, feared by intermediatelevel engineers, and used by expert engineers. Certain conditions are necessary to use these advanced measurement techniques, and simple test methods can reveal otherwise immeasurable timedomain characteristics.
Apparatus and background information
For sampled data systems, you need both a sampling system and a signal to sample. DSOs (digitalstorage oscilloscopes) have distinct advantages for the description and verification of sampling phenomena because of their crystalcontrolled switchedrange timebases, displays, and measurement cursors.
A few words of warning for those wishing to test these findings on their own equipment: You must disable any form of peak detection, which your equipment may refer to as envelope detection, maximumminimum mode, or glitch detection. Next, disable any advanced dotjoining features, such as sine or sinc interpolation; you need linear interpolation for these tests so that you can see the real sample points. Finally, disable any antialias filters that are available to the user. Equipment that does not allow users to disable any internal antialias filters is unsuitable for these tests or measurements.
Analogsampling oscilloscopes can help demonstrate the idea of correlated undersampling. They are relatively obscure devices for average engineers, partially because they can require significant skill to produce something other than a screen full of disconnected dots. Analogsamplingoscilloscope technology has existed for many decades and has achieved bandwidths in the gigahertz region. Analogsampling oscilloscopes are entirely different from DSOs. Analogsampling oscilloscopes take samples at a rate of perhaps 100,000 samples/sec and yet display waveforms with gigahertz and greater repetition rates. Clearly, analog sampling oscilloscopes take much less than one sample per waveform cycle, yet they manage to reconstruct timedomain signals with a defined accuracy.
How do analog sampling oscilloscopes achieve this accuracy (Figure 1)? The trigger event occurs several dozen nanoseconds ahead of the sample point and initiates a variable monostable that causes successive sample points to occur at an increasing distance from the trigger event. Thus the analog sampling oscilloscope can build up a stable display of a waveform, which is many times faster than the sample rate. For example, a modern, 50GHzbandwidth analog sampling oscilloscope can display 10GHzrepetitionrate signals using a 100,000sample/sec sampling rate. This ability is an example of correlated undersampling, because a trigger exists, and the sample point moves deterministically through the waveform.
DSOs employ a different technique. For a DSO sampling at 500M samples/sec, the sample points are 2 nsec apart. Thus, timebase speeds in which the screen displays more than 2 nsec use multiple points from one acquisition sweep. A randomsampling system addresses the desire to display the edge of the waveform without using an external delay line to get the advanced trigger. The DSO acquires a waveform at random, measures the time interval between the trigger signal and the internal timebase, and then figures out where to position the acquired point in a reconstruction of the waveform. This wellknown technique is called ETS (equivalent time sampling), and you can define it as a form of postcorrelated undersampling in which sample acquisition occurs prior to determination of its time position.
Trigger system
It is important to know what sort of trigger system your acquisition system is using. There are two distinct types. The first, and most common, system is the analog trigger. With this system, an analog signal feeds an analogtrigger comparator, and the level crossing a particular threshold generates a trigger pulse. Many variations on this theme exist, including pulse generation when the signal is outside a range or too small, but all systems based on analog signals are generically grouped as analog triggers.
Digitaltrigger systems, in contrast, are systems in which any trigger comparison takes place on the digital data stream that the main channel ADC creates. It is often difficult to notice any functional difference between these analog and digitaltrigger systems for a waveform that is within the range of the display. If you can't find the manual for your particular system, a few simple tests will enable you to tell whether the equipment has an analog trigger or a digital trigger.
Split a signal into two inputs of the equipment using a Tpiece or similar splitter. If you can shift the trigger channel trace completely off the display and still get a stable view of the waveform on the other channel, you have an analogtrigger system. Or, if the system has ETS timebase ranges, it has an analogtrigger system. If you set the triggerlevel markers to the middle of a known aliased waveform and the trace is steady, you have a digitaltrigger system; an analogtrigger system gives an apparently random trigger point on this test.
The distinction between these two types of trigger systems is important when you are dealing with aliases. Arguably, the analogtrigger system is best when you're trying to avoid aliases, and the digitaltrigger system is better when you are deliberately trying to create a stable alias.
Uncorrelated undersampling is popularly known as "aliasing"; the older term "beating" is still also used on occasion. An alarming number of engineers consider aliasing bad, dangerous, unacceptable, and not at all useful. For example, for some equipment, the manufacturer of the sampling system goes to the trouble of limiting the system bandwidth so that aliasing does not bother the users. It is a matter of popular experience that an alias of a sine wave looks pretty much like the original sine wave. What is perhaps less well known is that the alias of a square wave is a square wave—but not just any square wave. The alias is a timescaled replica of the original, if the alias conditions are correct. This point is very important.
The alias is more easily recognizable on a modern DSO than on one designed 15 or 20 years ago. The tendency now is to have an accurate verticaltriggerlevel marker displayed on the screen. Suppose the triggerlevel marker is centered on the trace, the trigger coupling is set to dc, and the trigger light is on, but the waveform is unsteady. This situation is a clue that aliasing is occurring. Of course you still have to check that you are triggering on that channel! Note that this clue applies only to analogtrigger systems. If the trigger is developed from a digital comparison against the acquired digital data, then the alias will appear in a stable position on the screen.
In ordinary use, you strictly avoid the alias, and if you find one, you change the conditions to make it go away. You might achieve this objective by increasing the sample rate, selecting some sort of peak detection mode, or filtering the incoming signal. You can also accomplish useful measurements by deliberately aliasing the signal. A few specific uses of aliasing follow. After seeing these applications, you may be more interested in the generalized theory behind the measurements (see sidebars "Definitions" and "Theorems").
Suppose a particular digitalacquisition system has a sampling rate of 10M samples/sec and a bandwidth of 5 MHz. You've measured the bandwidth by a sinusoidal envelopeamplitude method, the standard 3dBdown bandwidth point. You want to know what the pulse response is like, as this may be adversely affecting the measurements. You expect a 5MHz bandwidth system to have a rise time of around 70 nsec, as the approximation formula suggests:
The acquisition points are 100 nsec apart, so there is no chance of evaluating the pulse response of the system. Or is there? If you put a square wave with a repetition rate close to 10 MHz into the system, the resulting alias does not give a good representation of the pulse response. The square wave is just too fast for the bandwidth. The edge does not have a chance to settle out before the next transition. A 1MHz square wave is more appropriate in this case. The expected rise time of 70 nsec is then a smaller part of the cycle, and you can more clearly see the edges of the squarewave response. However, a 10Msample/sec sampling rate will not alias a 1MHz square wave.
Your options are to either turn down the sampling rate on the system, if possible, or to discard nine out of every 10 of the acquired points. This decimation process gives the desired sample rate of 1M samples/sec. Notice that you can do this decimation external to the sampling system, by transferring 10 times as much data as required, then selecting one of every 10 points by using, for example, a Visual Basic program. You need to apply a square wave that has a repetition frequency of slightly more than 1 MHz. If the squarewave frequency were exactly the same as the sampling frequency, you would get, in principle, a straight line. If the squarewave frequency is slightly lower than the sample rate, the alias is a timereversed and scaled version of the system pulse response. However, if the squarewave frequency is slightly higher than the sampling rate, you get the desired result.
Using a digital timebase on the sampling system and a synthesized signal generator set to 1.002 MHz, the nominal expected alias is about 2 kHz. You can achieve additional accuracy by measuring the actual alias frequency. A 2kHz signal has a repetition period of 500 µsec, which is 500 points at a 1Msample/sec acquisition rate. You measure the displayed frequency of the waveform as, for example, 2.05 kHz, but you know that the actual frequency is 1.002 MHz. Thus, you need to rescale time by the factor
Multiply the rise time of the aliased square wave by this scaling factor, and you have the rise time of the actual system. Suppose the rise time of the alias measures 35 µsec. You then calculate the rise time of the system using the equation
The way to solve these types of problems is to initially use a slower sample rate and a lower alias frequency. Then, you can increase the sample rate to 1M sample/sec so that the aliased edge has more points on it. This increase enhances the resolution of the measurement. In any case, the overshoot and shape of the pulse are always correct. Note that if you reduce the alias frequency, less than one complete alias cycle may appear on the 1Msample/sec range. If the equipment has an analogtrigger system, then you'll probably need more than one acquisition sweep to capture the rising edge. This task is easy when you do it manually. Just keep doing "singleshot" acquisitions until you get one with the rising slope roughly where you want it.
For this sort of test, a digitalcomparatortrigger system might actually be preferable, provided it can trigger on the displayed data. Note that if the system acquires at 10M samples/sec, the display may give decimated data at a 1Msample/sec rate, but the trigger comparison may still be done at 10M samples/sec. Unfortunately, manufacturers' data sheets probably don't provide this level of detail.
Sampletosample stability
A crystal ordinarily controls the timebase of a digitalacquisition system to a high degree of accuracy. Although ±100ppm absolute accuracy was the norm a few years ago, recent advances in crystal technology mean that timebase accuracies of ±25 ppm, ±10 ppm, and better are becoming more common. Novice engineers might therefore think that the time separation between two adjacent sample points is accurate to the published timebase specification. Consider this from a more practical point of view.
On an acquisition system with a sample rate of 500M samples/sec, the interval between dots is 2 nsec. Accuracy of ±25 ppm on this interval gives rise to an uncertainty of ±0.05 psec. This expectation is unrealistic for sampletosample jitter in any normal acquisition system. Realize that the timebase specification measures over some longer time interval—perhaps 10 to 100 msec long. Thus, the quoted accuracy is of an average interval and does not represent the sampletosample jitter due to phase noise in the master timebase oscillator and jitter in the logic gates.
Because sampletosample separation is not a specified, guaranteed, or even measured quantity for most acquisition systems, anyone making detailed time measurements may need to measure his or her equipment to establish the actual jitter. Highend DSOs may have characterized jitter performance, but they are out of the budget range of many users. One method of quantitative evaluation uses an alias.
In this approach, you first need a stable oscillator. The resulting measurement is going to include any jitter due to the phase noise in the oscillator. Thus it is essential that you use the best available generator. The next requirement is to minimize amplitude noise by using an insensitive range on the acquisition system and, correspondingly, a large amplitude from the oscillator. If there is any possibility of spurious components from the oscillator, you can improve noise by using a narrowbandtuned filter. It is simple enough to make up a passive LCR filter with 20dB extra insertion loss when the frequency is an octave (×2) away from the peak response. This technique improves any spurious noise sources on an already good oscillator.
As before, you are feeding a 1.002MHz signal into a 1Msample/sec acquisition system. The only difference this time is that you use a sinusoidal signal. A sinusoidal signal makes it easier to produce a clean, noisefree waveform. With a 2kHz alias frequency, you should not see any noise or distortion on the sinusoidal signal. As the alias frequency gets lower, however, the effects become much more noticeable. The sinusoid can become highly distorted when you reduce the alias frequency below 100 Hz. In fact, you may then be able to see the effect of analogtrigger breakthrough into the acquisition timebase clock. If your equipment has an analogtrigger system, it is therefore better on these tests to let the acquisition system "free run," rather than to use the signal as a trigger source.
Remember that the aliastoactualfrequency ratio must scale down in time any perturbation on the sinusoidal signal. If you fill the display with the 1kHz alias of a 1.001MHz signal by using a 100µsec/division timebase, then 0.1 division of jitter corresponds to a realtime jitter of
That is a huge amount! With a good analogsignal generator and a steady hand, you should be able to take the alias frequency down to 100 Hz or less. Notice that if the 100Hz alias is displayed across the whole screen by using an acquisition speed of 1 msec/division, 0.1 division of horizontal jitter corresponds to a realtime jitter of
Notice that the jitter as a ratio of the cycle time of the alias corresponds to the same amount of realtime jitter. You might, therefore, think of the jitter as being constant as you reduce the alias frequency in proportion to the timebase. This constancy does not happen in practice, however. As you reduce the timebase and alias, the jitter increases. At some point, the alias breaks up completely and is not even close to being sinusoidal or even periodic. The point at which alias breakup occurs is a powerful test of the stability of the internal master oscillator and the external oscillator. In fact, you can use alias breakup as a means of comparing different acquisition systems and oscillators.
So far, you have checked the timebasesampling interval to only 10nsec resolution. This resolution measurement is poor for a timebase that claims betterthan100ppm accuracy. You need to put in the lowfrequency alias but view it on the faster timebase. The trouble is that the signal will then appear as a straight line. You have to look for vertical deviation from this line, rather than for horizontal jitter. You previously established the vertical noise level of the equipment. Because you are on one of the least sensitive ranges and on a fast timebase, you should expect the noise to be around 1 LSB on an 8bit system.
Because the signal comprises six divisions point to point, the alias also comprises six divisions peak to peak. You either set the digital trigger to midscreen or, on an analogtriggered system, keep doing singleshot acquisitions until you get one near the middle of the screen. Suppose you see 0.1division pointtopoint noise on this straight line. This vertical deviation corresponds to a phase difference of
radians and, consequently, a time deviation of
This deviation then rescales by the aliastoactual frequency ratio, so that the alias frequency drops out of the equation. The timebase jitter is therefore
The best resolution available for the sample interval measurement in this situation is about 5 nsec, unless you resort to curvefitting techniques to more accurately estimate the jitter.
Although the resolution of this sort of test is relatively low, it will still show the weaknesses in multiphaseacquisition systems, such as CCDbased equipment and systems that interleave ADCs to get faster sample rates. Resolution of 5 nsec is not the best this system offers. You could take the amplitude jitter down to the leastsignificantbit level and extract a bit more resolution on the measurement. However, amplitude and time jitter become indistinguishable at this level. If the system had a 1Gsample/sec sample rate, then you should expect to be able to resolve down below the 5psec level by simple scaling of the previous results. However, when these numbers run down into the picosecond region with fivedigit resolution, you should be highly suspicious of the subfemtosecond results that some custom jittermeasurement software displays!
Definitions The basic sample rate is a sample rate that is very close to the signal frequency, ideally less than 5% away. Integer divisions of this sample rate are called decimated rates , even if the acquisition equipment can sample at only 1/10 the basic sample rate. Decimation by the integer (whole number) amount D means keeping only one sample from every block of D samples. The sample always remains at the same relative position in the block of D samples. Positive undersampling of a timedomain signal occurs when the repetition rate of the signal is slightly higher than the nearest integer multiple of the decimated (effective) sample rate. Negative undersampling has a signal frequency slightly lower than the nearest integer multiple of the decimated (effective) sampling rate. Given a signal frequency F_{SIGNAL} and a sample frequency F_{SAMPLE} , you can state the definition of undersampling mathematically as F_{SAMPLE} &2´F_{SIGNAL} . The "mod" function is defined as In other words, the mod function gives the remainder when A divides into K as many whole number of times as possible. Using this function, positive undersampling occurs when and negative undersampling occurs when

Theorems Philosophers and physicists have observed the general concept of alias phenomena over the ages. Similarly, cinema audiences and TV viewers all have seen the wheels on stagecoaches apparently rotating backwards and propellers on airplanes apparently slowing down, then going backwards. The main text of this article collates these aliasing results into a simple and quantitative form, with specific application to the acquisition of data. The following theorems provide a generalized background for the examples. Frequencydomain alias theorem: For a modulated signal (frequency domain), ideal undersampling loses the carrierfrequency information but does not lose any of the modulation information, provided that the modulation does not extend out from the carrier by more than half the sample frequency. Alias theorem A: If the signal frequency is within 5% of the basic sample rate, there will be at least 20 samples per cycle on the alias. If the signal is within 1%, there will be at least 100 samples per cycle, and so on. In general, a signal within X% of the basic sample rate gives at least 100/X samples per cycle on the alias. Alias theorem B: Decimation by the integer amount D requires that the signal be closer to an integer multiple of the decimated sample rate to get the same number of samples per cycle on the alias. Specifically, being within X% of the nearest integer multiple of the decimated rate gives at least samples per cycle. Alias theorem B is, therefore, a generalization of Alias theorem A. Alias theorem 1: For an ideal repetitive timedomain alias, timerelated features on the alias, such as rise time, scale by the ratio of the repetition frequency of the alias to the repetition frequency of the original waveform. Amplituderelated measures, such as overshoot and peaktopeak aberration, are unaffected by ideal undersampling. Alias theorem 2: You can express the number of samples on each cycle of the alias mathematically as Alias theorem 3: For a repetitive timedomain signal, ideal positive undersampling loses the repetitionfrequency information but does not materially change the shape of the signal, provided that there are at least 10 points on the aliased rise time and that waveform artifacts are not smaller than 1/5 the rise time. Ideal negative undersampling gives a timereversed but otherwise correct alias (Figure A). By using these alias theorems, you can make some previously impossible measurements with defined accuracy. Also, you can quantitatively evaluate the sampletosample jitter of acquisition systems. Rather than being a nuisance, the alias can be a useful tool for sampling systems. 
Author info
Leslie Green has a bachelor's degree in electrical and electronic engineering from Imperial College (London). He has 20 years of experience working with testandmeasurement equipment, the last 15 years of which he spent designing DSOs for GouldNicolet Technologies in the United Kingdom.
REFERENCE
1. Nyquist, Harold, "Certain topics in telegraph transmission theory," Transactions of the AIEE #47, February 1928, pg 617 to 644.
System level design and integration challenges with multiple ADCs on single chip
Understanding the basics of setup and hold time
Product Howto: Digital isolators offer easytouse isolated USB option
Managing noise in the signal chain, Part 2: Noise and distortion in data converters
War of currents: Tesla vs Edison
Simple reversepolarityprotection circuit has no voltage drop
Control an LM317T with a PWM signal
Start with the right op amp when driving SAR ADCs
Currently no items